Thank you for the great suggestions. I did troubleshoot a little more, the
water chiller seems to work but when unplugging the water chiller at the
ICP MS outlet and capping the end to the water chiller and turning on the
water flow (via Start Plasma) nothing comes out of the ICP MS "Out" but it
should right? So would this suggest that the ICP is plugged? Please let me
know if I am wrong I will try to reverse the flow and see if whatever is
plugging it can be loosen up.
On Wed, Jun 20, 2018 at 9:14 AM, Walaa Abdulatif <[log in to unmask]>
> Hello Birgit,
> This model ans SW includes a log for the plazma events. There will be a
> listed informative info.
> Water no flow is normal scence plazma not maturely on for this model.
> It happens some times valve late on and water not sensed flowing on time,
> then plazma swoched of by controller.
> Let me know the enent errors from your sw to get more glow.
> Walaa wadan
> Icp specilist
> Nilebasin sci
> Riyadh, ksa
> [log in to unmask]
> On Wed, Jun 20, 2018, 7:34 PM Birgit Hagedorn <
> [log in to unmask]>
> > Hello
> > I am setting up a used and refurbished Elan 9000 and new at this
> > brand. The plasma ignition sequence is shutting down and the
> > say "low water flow". I am not sure if his is a water chiller problem or
> > something in the ICP and was wondering if anybody has some guidance and
> > experience with Elan 9000 and how to find out why the water flow is low?
> > Thank you,
> > Birgit
> > --
> > Birgit Hagedorn, PhD
> > CEO, OWNER
> > *Sustainable Earth Research LLC*
> > 2200 Alder Drive
> > Anchorage, AK 99508
> > phone: 907.351.5362
> > email: [log in to unmask]
> > www.searchlc.com
> > check for new training classes:
> > www.environmentalsampler.com
Birgit Hagedorn, PhD
*Sustainable Earth Research LLC*
2200 Alder Drive
Anchorage, AK 99508
email: [log in to unmask]
check for new training classes: