Apologies, but the link in my previous email sees to be pointing to the
I hope this works.
On 29/01/2010 15:35, Jason Day wrote:
> Hi Plasmachem Friends-
> Very embarrassing, during a routine lens cleaning, I noticed the small
> screw on the chamber lid near the hinge was in bad shape so I took the
> whole lid off to sort out that screw. That job is done.
> But, during this time, I forgot how the wires go back together. I
> can't find my notes or any photos. I know there are around here
> Can anyone help to tell me which wire goes where? I'm talking about
> the two push on gold connectors that attach to the lid of the DRC/lens
> chamber. Does the one closest to the plasma go to the pin closest to
> the front of the instrument? I didn't want to hook it up wrong and
> then take it apart again as the screws have seen better days.
> We have a DRCII from 2002. There is a photo at the link below - I hope
> it gets through!
Department of Earth Sciences, University of Cambridge
Cambridge, CB2 3EQ, United Kingdom
Tel:(office) +44 (0)1223 765 282
Mobile 07889 408 757
Fax:(dept) +44 (0)1223 333 450