We do this repeatedly.
In your TERM and ERRTERM, the "Type" column you should have Keyword and in the "Data File" column you should have StandBy
One more trick is to make sure you have no open windows in the ICP_MS Top window and the pumps will stop too.
Email me back if you have more questions,
[log in to unmask]
State Hygienic Laboratory at the University of Iowa
2220 South Ankeny BLVD
Ankeny, Iowa 50023
From: PLASMACHEM-L: Analytical Chem.(ICP's, DCP's, MIP's). [[log in to unmask]] On Behalf Of Jay Tompkins [[log in to unmask]]
Sent: Friday, December 03, 2010 2:11 PM
To: [log in to unmask]
Subject: overnight runs with Agilent 7500a
A client has asked if we could run samples overnight. We have never done this although I was told that in chemstation there is a command that can be added to a sequence that will turn off the plasma but everything else keeps running as far as I know. Does anyone have any ideas on overnight runs?
ABC Laboratories, Inc., 4780 Discovery Drive, Columbia MO 65201.
Tel: 573-777-6012; Fax: 573-777-6090
Integrated Chemistry and Biology Services for Product Development