Is there any user group or forum for ICAP users to discuss such
specific software/hardware issues?
On Fri, Sep 23, 2011 at 4:55 AM, Dhinesh Asogan <[log in to unmask]> wrote:
> Dear Abe,
> DigiLAZ III has a feature to enable a constant contact closure signal to allow the iCAP to run multiple replicates through its generic trigger capability. In the latest version of the software, there is an option to 'Enable Trigger for Entire Sample' under the triggering tab, which will set the contact closure to high during the ablation runs, and will shut off between patterns.
> We provide a cable to connect the LSX systems to the DIN connector on the front of the iCAP. iTEVA then needs to be configured for Laser TRA under Tools > Instrument Options > Sample Introduction tab: select the 'Custom' accessory, disable the drain sensor, set the Input Trigger to 'Active High' and the Output Trigger to 'Active Low'.
> I tried calling you this morning, but the number you provided seems to be a Portsmouth landline number and is missing a digit. UK mobiles usually start with 7? I tried adding the 7, but that didn't work either. I notice also you didn't include your affiliation, can you please let me know and maybe I can find you offline that way? Alternatively, please feel free to call me if you need further advice.
> Kind Regards,
> Dhinesh Asogan
> Laser Ablation Product Specialist
> CETAC Technologies
> 9 Alderman Walk
> SS17 8DA
> United Kingdom
> Tel: +44 1375 404 742
> Mob: +44 7747 843 061
> The content of this message is the property of CETAC Technologies and is proprietary. The only authorized audience are the original recipients of the message, or member of the original recipient's organization. Use of the content of this message is pursuant to permission by CETAC Technologies. All other uses of this information are expressly forbidden.
> -----Original Message-----
> From: PLASMACHEM-L: Analytical Chem.(ICP's, DCP's, MIP's). [mailto:[log in to unmask]] On Behalf Of Abe Morton
> Sent: Friday, September 23, 2011 2:24 AM
> To: [log in to unmask]
> Subject: Thermo ICAP Laser Ablation transient analysis
> Does anyone have a sample Thermo ICAP method for connecting a Cetac laser ablation system with the ICAP. It seems this ICP-OES was designed for a newwave LA software setup. We have Digilaz III software and we are trying to do a 50 multi-line scan with each line being a separate icap sample analysis.
> Abe Morton,
> Analytical Chemist,
> Mobile +44-23-587-84-19
New York School of Medicine