Is your Meo software performing a purge on the sample chamber prior to
starting the experiment? You will need to add that time onto your PlasmaLab
timing as well (duration in the Sample List). For what it's worth, I took
off the uptake time, reduced the laser warm-up to 10sec and reduced the
purge to 2s to speed things up.
The other issue could be the triggering in Meo. Check your Sync Out (in Run
Experiment) ... the 'Active during Entire Experiment' should be selected and
'Active Closed' selected under Polarity. For Trigger In, 'Don't wait for
Trigger' should be selected.
Oh also, make sure that you select 'All Laser Patterns' in Run Experiment.
- Show quoted text -
Quantitative Bioelemental Imaging Center (QBIC)
Silverman Hall 4578