Are you using autosampler?
It can have a clog....
For nebulizer do not use ultrasound!!!
You can use a serynge to push some possíble clog... Triton 0.1 % and nitric
Em 01/03/2017 12:32 PM, "Michael Ruhlman" <[log in to unmask]> escreveu:
> It definitely looks like the nebulizer was causing some problems. I put
> it under the microscope and am pretty sure I see build up.
> I am going to try a Sea Spray nebulizer to see if that will help some.
> Does anyone have experience with a pre-treatment of the nebulizer or a wash
> solution that could help clear any protein build up?